Jual Davis AeroCone Rain Collector

Jual Stasiun Pemantau Curah Hujan Indonesia

The AeroCone rain collector reduces wind-induced errors by providing a more aerodynamic path around the collector.
Features

• Contours streamline the airflow and reduce flow distortions
• Interior walls that reduce splash out during high rain conditions
• Debris screen locks into place
• Jual Stasiun Pemantau Curah Hujan Indonesia

This rain collector can be used with Vantage Pro2™ weather stations, including Vantage Pro2, Vantage Pro2 Plus, and the Anemometer/Sensor Transmitter Kit.
Note: Model number 6463 is factory-calibrated to take rain measurements in 0.01 inches. The metric version, 6463M, comes factory-calibrated to take measurements in 0.2 mm.

Jual Stasiun Pemantau Curah Hujan Indonesia

Specifications Sensor Type. . . . . . . . . . . . . . . . . . . . . . Tipping bucket with magnetic switch
Output . . . . . . . . . . . . . . . . . . . . . . . . . . Contact closure
Attached Cable Length. . . . . . . . . . . . . . 40’ (12 m)
Cable Type. . . . . . . . . . . . . . . . . . . . . . . 4-conductor, 26 AWG
Connector . . . . . . . . . . . . . . . . . . . . . . . Modular connector (RJ-11)
Recommended Max.Cable Length . . . . 900’ (270 m)
Housing Material . . . . . . . . . . . . . . . . . . UV-stabilized ABS plastic

Dimensions
Rain Collector and base. . . . . . . . . . 8.75" diameter x 9.5" high (22.2 cm diameter x 24 cm high)
Collection Area. . . . . . . . . . . . . . . . . 33.2 in2 (214 cm2)

Range
Daily Rainfall . . . . . . . . . . . . . . . . . . . 0.00” to 99.99” (0.0 mm to 999.8 mm)
Total Rainfall . . . . . . . . . . . . . . . . . . . 0.00” to 199.99” (0.0 mm to 6553 mm)
Accuracy . . . . . . . . . . . . . . . . . . . . . . . . For rain rates up to 2"/hr (50 mm/hr): ±4% of total or +0.01” (0.2mm) (0.01" = one tip of the bucket), whichever is greater. For rain rates from 2"/hr (50 mm/hr) to 4"/hr (100 mm/hr): ±5% of total or +0.01” (0.2mm) (0.01" = one tip of the bucket), whichever is greater.
Update Interval. . . . . . . . . . . . . . . . . . . . 20-24 seconds

Input/Output Connections
Red . . . . . . . . . . . . . . . . . . . . . . . . . . Switch terminal
Green & Yellow . . . . . . . . . . . . . . . . . Switch terminal

If you want a more detail specification, feel free to contact Us or email Us for further information.
Greet from Kenzo, Mealabs Environment Indonesia's Representative

Jual Stasiun Pemantau Curah Hujan Indonesia

If you want a more detail Accessories, feel free to contact Us or email Us for further information.
Greet from Kenzo, Mealabs Environment Indonesia's Representative